The importance of measuring the pH of pure water
Measuring the pH of pure water is a fast and reliable way to detect contamination during the manufacture or distribution of pure water. The introduction of gaseous impurities, such as air or carbon dioxide, will alter the pH of the clean water, indicating the infiltration of air or gas into the process line. The depletion of resin beds employed for the preparation or refinement of clean water can lead to the introduction of impurities, resulting in a pH alteration. The resin beds commonly consist of cation and/or anion exchange resins in either the hydronium ion form (cation exchange) or the hydroxide ion form (anion exchange). Conductivity measures can effectively detect ionic leakage, while changes in pH measurements can provide diagnostic information about the depleted resin. If the pH rises, it indicates that the cation resin bed is being depleted; conversely, if the pH lowers, it suggests that the anion resin bed is becoming depleted.
Boilers necessitate the use of pure water to minimize the formation of scale and the transfer of contaminants in steam. Corrosion can happen when the pH level goes beyond the permissible limits, which vary depending on the metallurgies present in the steam cycle.
For optimal product quality, semiconductors necessitate the use of highly purified water. With the advancement of nanotechnology, contaminants act as significant obstacles for microcircuits, potentially leading to unwanted short circuits. The measurement of pH can also aid in the management of corrosion in these items.
International pharmaceutical quality standards mandate the highest level of water purity to ensure the safety and efficacy of medications. The pH range of 5-7 at 25°C is mandated by the USP 23 standards.
To summarize, measuring the pH of pure water can provide assurance that the water being utilized maintains its maximum level of purity for the intended purpose. Factors to Consider while Taking Measurements Implementing preventive actions is essential to guarantee the precision of pH measurement in clean water. It is essential to examine the following factors: Infiltration of atmospheric carbon dioxide (CO2) When clean or pure water, usually with a pH of 7, is exposed to the atmosphere, it will absorb carbon dioxide (CO2) until it reaches a state of equilibrium. Carbon dioxide (CO2) causes acidification of the sample, resulting in a decrease in pH. The pH of pure water can decrease to a minimum value of 6.2, depending on the temperature and pressure conditions. An optimal resolution to this issue entails directing a sample stream via a compact, hermetically sealed sample chamber housing a pH electrode assembly. One alternate approach is to raise the temperature of the water sample in order to remove the absorbed CO2 right before testing the pH. The reboiler sample system is frequently employed in power plants to eliminate CO2 from clean water. Avoid transporting grab samples to a lab meter, as the presence of ambient CO2 can compromise the integrity of the sample. Nevertheless, the laboratory meter can be utilized at the sample station even when the grab sample is excessively full. In this approach, the sensor is positioned within a container, specifically a beaker, containing the material that needs to be quantified. The sample must be introduced into the beaker by submerging a tube to the bottom of the beaker. To prevent the infiltration of CO2 from the surrounding air, it is necessary to intentionally allow the sample to overflow the beaker. It is necessary to consider temperature correction for pure water.
Please contact us to discuss your application